- Joined Aug 27, 2009
All inbuilt AWG's offer only limited amplitude yet X Plus AWG does have the 50 MHz claim to fame.Enjoy! I got my Siglent 2104X+ 4ch 8 or 10 bit 100MHz a year ago for slightly less (before tax) with the 50MHz built in AWG plus it does neat things like Bode plots. Only thing I wish it would have is a higher peak to peak output on its AWG.
Correct, SDG1032X can be convinced to become SDG1062X and few in this class can match either models risetime.But I also have their 30MHz bench AWG that does 20Vpp and there are signal amplifiers if needed. Not doing anything that needs more than 30MHz but it can always be upgraded (or hacked).
There are DIY LA probe options for X Plus and as they are a passive probe assembly connecting into a PCI-e socket, none of the LA assembly is proprietary however the 10 way flat ribbon coax is very difficult to find.The Siglent does have the ability to do 16 channel logic IF you buy their wiring harness and Upgrade the firmware for it. Maybe one of these days...
Edit/correction made.True but he's no fly by night BS video guy, his opinions hold sway on test equipment because he's paid the dues of being around for a while and has the equipment and some expertise to back it up.
Showoff!View attachment 303636
View attachment 303637
The VESA mount blocks some of the air-flow so I fixed that later with longer screws and plastic standoffs.
View attachment 303638
First boot with all the inputs and probes for testing.
View attachment 303639
A mess of probes for two scopes.
View attachment 303645
View attachment 303641
Spacers added to the VESA mount to improve air-flow.
View attachment 303642
External HDMI monitor display.
View attachment 303643
Looks very good.
1 All interrupt triggers toggle the gpio pin.
2. Main processing run is high logic level. Every super-loop iteration sets the gpio pin low on exit and back high when it starts at the beginning.
3 and 4 are I/O signals.
Looking for program bottlenecks caused by interrupts or mainline programming from variations (gaps and changes in toggle densities) in the traces.
View attachment 303646
Everything works but there are some GUI and function quirks normally seen with 1.0 firmware. It's too early to really evaluate total functionality but I'm happy with it so far.
View attachment 303644
|Thread starter||Similar threads||Forum||Replies||Date|
|How 10bit SAR ADC achieves 12bit resolution||Analog & Mixed-Signal Design||4|
|M||STM32L471 -ADC clock frequency(fADC)=80 MHz, Sampling rate for FAST channels(fs)=5.33Msps for 12bit||Analog & Mixed-Signal Design||5|
|F||A Low-Pass filter for 12bit ADC . How to calculate?||Microcontrollers||7|
|SPI routine for interfacing serial 12bit adc with a philips 89c51RD2 Mcu||Programming & Languages||0|
|M||Pls recommend DAC 12bit i2c or spi||Analog & Mixed-Signal Design||1|
|How 10bit SAR ADC achieves 12bit resolution|
|STM32L471 -ADC clock frequency(fADC)=80 MHz, Sampling rate for FAST channels(fs)=5.33Msps for 12bit|
|A Low-Pass filter for 12bit ADC . How to calculate?|
|SPI routine for interfacing serial 12bit adc with a philips 89c51RD2 Mcu|
|Pls recommend DAC 12bit i2c or spi|