Just a word of caution. Neutral may not be at ground potential at your location and if your scope's case is grounded then high currents could flow.Originally posted by nomurphy@Aug 9 2005, 04:53 PM
Please don't connect your probe directly to the AC line, you could easily blow your probe or the input stage to your scope. Using a step-down transformer is best, but barring that, attach a 470K (top/hot) and 22K (bottom/neutral) in series and **CAREFULLY** (one handed) measure across the 22K to neutral.
And, as noted before, your scope should be plugged into the same AC source and properly polarized (hopefully three-pronged AC cord with ground).
|Thread starter||Similar threads||Forum||Replies||Date|
|Measuring current using a Shunt Resistor & an Oscilloscope||Test & Measurement||11|
|R||Best ways to record data on interference pattern using Tektronix DPO 5204B oscilloscope?||Test & Measurement||3|
|A||Current Measurement using oscilloscope||Test & Measurement||28|
|Measuring bandwidth of OWON VDS3102 oscilloscope using "Trigger Out" signal from "MULTI" connector||Test & Measurement||10|
|U||using oscilloscope crt+power supply||Power Electronics||41|
|Measuring current using a Shunt Resistor & an Oscilloscope|
|Best ways to record data on interference pattern using Tektronix DPO 5204B oscilloscope?|
|Current Measurement using oscilloscope|
|Measuring bandwidth of OWON VDS3102 oscilloscope using "Trigger Out" signal from "MULTI" connector|
|using oscilloscope crt+power supply|