Good to hear you got it working again.
Contacts and capacitors can be a hard time to trace.
I hope the calibration is still OK for you.
|Thread starter||Similar threads||Forum||Replies||Date|
|calibration passive probe of scope||Test & Measurement Forum||4|
|Telequipment D61a Triggering Issue||Technical Repair||7|
|P||Scope Telequipment D61A;horizontal position fault||General Electronics Chat||232|
|T||telequipment d75 oscilloscope (failing afther some time)||General Electronics Chat||1|
|O||Telequipment Oscilloscope D75||Test & Measurement Forum||9|