I am trying to verify that an electric meter pulser is putting out pulses correctly. My literature says that it puts out a 500 milisecond pulse, dry contact closure, it is being supplied with 10vdc on the common. I believe it is caled a Form A pulse?? My concern is that my Digital multi meter is not seeing that quick of a pulse and that i am simply missing them, due to the DMM not "refreshing" quick enough. I have no clue though, could just be a broken pulser. Any input is appreciated......unless its disparaging....nobody likes that kind.