# 3 phase induction motor Lab

Discussion in 'Homework Help' started by Jess_88, Oct 4, 2011.

1. ### Jess_88 Thread Starter Member

Apr 29, 2011
174
1
Hey guys

I'm having some trouble with what I need to determine from the measurements made in my lab.

I'll copy all of my lab procedure and parameters to avoid as much confusion as possible.

The objective of this experiment is to investigate various methods of induction motor speed control. How the standard motor characteristics are affected by the different methods of speed control, so as to allow informed decisions on the best type of control for a given application The Experiment includes:
 Variable stator voltage control
 Rotor resistance control
 Frequency control

VARIABLE STATOR SUPPLY VOLT AGE (DEL T A CONNECTED)
1) Ensure the cage induction motor is Delta connected and the dynamometer is cumulatively compounded with minimum excitation.

2) Set the panel autotransformer to the 440 V tap. Switch the motor on and record the no-load speed and line current.

3) Set the dynamometer load to 10 N and record the induction motor line current and speed. (Note: Torque arm length is 0.25m)

4) Switch off the motor. Set the panel autotransformer to the next voltage level tap down and repeat steps 1 & 2

5) Repeat the above steps for successive voltage level taps down to the 220V tap.

6) Plot on a common graph Speed V Voltage for Delta connections Loaded and on No Load.

MY RESULTS

VARIABLE STATOR SUPPLY VOLTAGE (STAR CONNECTED)
1)Ensure the cage induction motor is Star connected and the dynamometer is cumulatively compounded with minimum excitation.

2) Set the panel autotransformer to the 440 V tap. Switch the motor on and record the no-load speed and line current.

3) Set the dynamometer load to 10 N and record the induction motor line current and speed. (Note: Torque arm length is 0.25m)

4) Switch off the motor. Set the panel autotransformer to the next voltage level tap down and repeat steps 1 and 2

5) Repeat the above three steps for successive voltage level taps down to the 220V tap.

6) Plot on a the same common graph Speed V Voltage for Star connections Loaded and on No Load so that a comparison can be made with the Delta connection.

My results

note- the results in red i think my be incorrect... maybe

VARIABLE ROTOR RESISTANCE. (DELTA CONNECTED)
1) Ensure the Wound Rotor Induction motor is Delta connected and the dynamometer is cumulatively compounded with minimum excitation.

2) Adjust the rotor resistance set to the first position. Switch the motor on and record the no-load speed and line current. Repeat for all rotor resistance positions.

3) Set the dynamometer load to 20 N and record the induction motor line current and speed. Repeat for all rotor resistance positions.

4) Plot a graph of resistor position Vs Speed for both no load and load and comment on the characteristic produced.

My results

VARIABLE VOLTAGE VARIABLE FREQUENCY CONTROL
1) Ensure the cage induction motor is Star connected.

2) Connect the inverter input terminals to the three phase supply on the panel and its output terminals to the motor with the power analyser set to measure
frequency. This is on voltage current and hertz range with the voltage inputs set on motor line to line voltage.

3) Vary the inverter frequency from 25 Hz to 60 Hz in steps of 5 Hz. Record frequency, speed, voltage and line current for each step.

4) Calculate the synchronous speed and slip for each test.

5) Plot a graph of Frequency Vs Speed of the motor and comment on the result.

So here is my confusion. Can I really determine anything from plotting Voltage against speed? Should I be tying to determine the equivalent circuit???
I am guessing I need to determine the torque induced for each test, then plot that agains Ns to determine the motor class. Then I can determine the motor application right?

How can I calculate the slip without the rated speed???

Let me know if you want to see my graphs.

thanks guys

2. ### Jess_88 Thread Starter Member

Apr 29, 2011
174
1
My results from VARIABLE VOLTAGE VARIABLE FREQUENCY CONTROL

note - sync speed was calculated from Ns = (120*f)/4

3. ### strantor AAC Fanatic!

Oct 3, 2010
4,302
1,988

Jess_88 likes this.
4. ### t_n_k AAC Fanatic!

Mar 6, 2009
5,448
782
Keep in mind that in Australia the mains frequency is 50Hz.

Jess_88 likes this.
5. ### strantor AAC Fanatic!

Oct 3, 2010
4,302
1,988
doh! missed that

EDIT: despite that oversight, he's still running above synchronous speed at 50Hz in variable frequency test, so something still doesn't add up.

EDIT 2; should still double RPM when in star

Jess_88 likes this.
6. ### Jess_88 Thread Starter Member

Apr 29, 2011
174
1
Is difference in expected results maybe because the torque arm length is 0.25m for all tests but the variable voltage variable frequency control test???

7. ### strantor AAC Fanatic!

Oct 3, 2010
4,302
1,988
No, I thought 1.1, 1.3, and 3.1 were no-load tests. if 1.1 & 1.3 were no-load, and 3.1 was with a load, then that means the error is even greater than I thought.

Jess_88 likes this.
8. ### Jess_88 Thread Starter Member

Apr 29, 2011
174
1
sorry, my mistake (I feel silly). Yeah data from table 1.1, 1.3 and 2.1 are no load. As for Table 3.1... I'm a bit confused (I can't remember applying a load). Shouldn't the variable voltage variable frequency control test be done without a load?
goodness me... I feel like I could be in trouble. I was sure I was making my measurements correctly

9. ### strantor AAC Fanatic!

Oct 3, 2010
4,302
1,988
well if you did all the other tests with & without a load, I would expect you would do the variable voltage variable frequency test both with and without a load as well. So are you missing a table of data?

I'm slightly confused myself now; I think the key (or part of it) lies in the voltage output of the transformer. I was using the numbers from the variable frequency variable voltage test (444V @ 60Hz) to determine what your RPM would be in the other tests, but then learned that those were 50Hz tests, so now I don't have much to go on. I strongly suspect that you flubbed some measurements in the variable frequency variable voltage test.

Jess_88 likes this.
10. ### GetDeviceInfo Senior Member

Jun 7, 2009
1,571
230
I personally don't see any problems with your results. Yes, the graphs are a quick visual reference to the spectrum.

why would rpm double going from delta to star?

I agree that a loaded test should have been performed for variable frequency.

If I'm not mistaken, the intent of the exercise is to realize speed regulation over the varying methods.

Jess_88 likes this.
11. ### strantor AAC Fanatic!

Oct 3, 2010
4,302
1,988
Well, for his variable voltage variable frequency test, the motor is spinning at or above synchronous speed. that's the main problem I see.

as far as star Vs delta, I had that backwards. going from star to delta should have doubled (or significantly increased) the speed.

Jess_88 likes this.
12. ### GetDeviceInfo Senior Member

Jun 7, 2009
1,571
230
I read that as variables in the procedures, but the overall intent (speed regulation) is demonstrated.

Delta or wye, the pole count is the same, there will be no change in speed, however, thier will be an implication in speed regulation at some loading level as the available power is a factor of 3

Jess_88 likes this.
13. ### strantor AAC Fanatic!

Oct 3, 2010
4,302
1,988
ah yes, I stand corrected. I was thinking of wye/delta starters like this one, and of the wiring diagrams in the 9 lead 230/460V motors for HV/LV, of how when you wire for wye, you split the voltage across 2 windings, vise sending it only through 1. 440V across the winding in delta would be 440(1/(3^.5)) = 254V across the winding in wye. I was thinking that would cut the speed in half, but you are right, unloaded, the speed would be the same.

Sorry for the confusion.

Jess_88 likes this.
14. ### Jess_88 Thread Starter Member

Apr 29, 2011
174
1
ok so assuming everything was measured correctly. I'm still a bit confused as to what parameters I need to determine. What can I determine plotting Voltage against rotation speed? what dose the Nsync actually tell me?

I'm guessing I will be needing the slip to determine something for the variable frequency test??? Would I calculate it by

Nslip = Nsync - speed

s = Nslip/Nsync

Can anyone tell me what I should be trying to fined in each test? I don't really know what direction to go with

thanks guys

15. ### strantor AAC Fanatic!

Oct 3, 2010
4,302
1,988
I think you're going about it backwards. You don't determine an application based off a table of measurements. You start out with an application, and based off these tables of data that you collected, you decide which configuration is best. I'm with GetDeviceInfo on this; I don't think you are supposed to be "solving" for any specific numbers. If they didn't give you the *given application, then I think the exercise is just to familiarize you with the different types of controls available, and stimulate you to think what they might be used for.

Jess_88 likes this.
16. ### Jess_88 Thread Starter Member

Apr 29, 2011
174
1
ah ok, thanks. Much appreciated